V 300

COMPACT
MICROSCOPE

FOR EDUCATION AND LABORATORY

Microscope with solid cast stand, equipped
with coaxial coarse and fine focusing with incorporated halogen lamp 6 V / 20
W, continuous intensity control. Mechanical stage 130 x 130 mm. Condenser
adjustable in height for precise Köhler
illumination, aperture diaphragm, filterholder.
Quadruple revolving nosepiece. Pair of eyepieces WF 10x/18. All routine
investigation methods such as brightfield, darkfield, phasecontrast and
polarization can be performed with the V 300 microscope.

MEDICUS HF AND PH FOR THE MEDICAL PRACTITIONER
FOR INVESTIGATIONS IN BRIGHTFIELD, RESPECTIVELY IN BRIGHTFIELD AND PHASE
CONTRAST

Binocular microscope for the medical practitioner with
incorporated halogen illumination 6 V / 20 W, with continuous intensity
control. Stand with coarse and fine focusing, condenser NA 1,25/0,6. Quadruple revolving nosepiece with achromatic
objectives AC 10/0,25, AC 40/0,65 and AC 100/ 1,25
(Oil). Pair of eyepieces WF 10x. Microscope MEDICUS PH for brightfield and phasecontrast
with objectives A Ph 40/0,65 and insertable light
annulus diaphragm for the condenser

Vb 365 FOR BRIGHTFIELD INVESTIGATIONS

Binocular laboratory microscope with 4 brightfield
objectives

A 4/0,10

A 10/0,25

A 40/0,65

A 100/1,25

and condenser
NA 1,25/0,6. Pair of eyepieces WF 10x/18.

Vb 365 FOR ORIENTATING INVESTIGATIONS IN POLARIZED LIGHT FOR
DETECTION OF CRYSTALS
IN PLANT CELLS AND MINERALS

Same equipment as Vb 365, but with
polarizing equipment.

Vb 365 WILOZYT FOR INVESTIGATIONS IN BRIGHTFIELD, DARKFIELD AND
PHASECONTRAST

Binocular laboratory microscope, equipped with

objectives A 10/0,25 (brightfield)

A Ph 20/0,35 (brightfield + phase)

A Ph 40/0,65 (brightfield + phase)

with combination condenser NA 0,9 for brightfield,
darkfield

and phasecontrast. Pair of eyepieces WF 10x/18.