THE MODULAR SYSTEM H 600

The modular design of the H 600 makes it a versatile
instrument suitable for a wide variety of applications. Routine  investigation methods such as brightfield, phase contrast, darkfield
and polarisation are all possible through easy
exchange of components. Reflected light, fluorescence microscopy,
photographic and video documentation are optional additions.

H 600 WILOCYT

FOR THE GYNAECOLOGY LABORATORY (FOR STAINED AND
UNSTAINED CELL PREPARATIONS)

Binocular laboratory microscope with variable halogen
lighting 6 V/30 W, quadruple reversed nosepiece, allowing easier access to
the specimen. Widefield eyepieces WF 10 x (18).

Objectives: A 10x/0.25

APh 20x/0.35

APh 40x/0.65

A 100x/1.25 (Oil)

Condenser NA 0.9 for brightfield,
darkfield and phase contrast (up to 40 x), filterslide

H 600 PHACO

As H 600 Wilocyt, but with
quintuple nosepiece and five semiplanachromat phase
contrast objectives, for unstained specimen preparations examined under phase
contrast.

H 600 WILO-PRAX

FOR THE GENERAL PRACTITIONER

lighting 6 V/30 W, quadruple reversed nosepiece, allowing access to the
specimen.

Widefield eyepieces
WF 10 x (18).

Objectives: A 4x/0.10

A 10x/0.25

A 40x/0.65

A 100x/1.25 (Oil)

Condenser NA 0,9 for brightfield, filterslide.

H 600 PL
5

THE INSTRUMENT FOR MORE DEMANDING USE

As H 600 Wilo-Prax, but with
quintuple nosepiece and five planachromat
objectives for examination of stained specimens in histology and cytology.
Recommended for good quality photographic documentation.

ACCESSORIES

OBJECTIVES

Ranges of
objectives for transmitted light investigations:

– Achromatic

– Semiplan achromatic

– Plan-C-achromatic

– Plan achromatic

– Phase contrast achromatic and semiplanachromatic

– Infinity corrected

EYEPIECES

Widefield eyepieces,
standard and for spectacle wearers are available in the following
magnifications: 10 x, 12,5 x.

For measuring eyepieces fitted with a graticule
are available.

BINOCULAR PHOTOTUBES

Binocular phototube 70/30

Light is split with 70% to the phototube and attached device

eg video
camera, 30 % to the eyepieces enabling simultaneous observation.

Binocular phototube 100/100

100 % of the light is directed either to the eyepieces or the
phototube. This system is most useful for low light intensity situations, as
all the available light is directed to the photographic device. Video cameras
are fitted to the phototube by means of a C mount.

DISCUSSION TUBE DB 2

For the simultaneous viewing of a specimen by two observers
a discussion tube equipped with a pointer is available.