REFLECTED
FLUORESCENCE LIGHT MICROSCOPE FOR ROUTINE INVESTIGATIONS

The H
600 models can be fitted with a reflected light fluorescence illuminator for
fluorescence light microscopy. The advantage is that transmitted light phase
contrast and reflected fluorescence microscopy can be combined for further
versatility. Our product range includes all the standard filterblocks
currently in use and two filterblocks can be
mounted on the filterset holder, allowing rapid
switching between different stainings

Binocular
laboratory microscope with variable transmitted halogen lighting 6 V/30 W,
HBO 50 W or 100 W, high pressure mercury lamps, approved for fluorescence
microscopy. Mechanical stage and quadruple reversed nosepiece, allowing
easier access to the specimen, optional quintuple nosepiece available.

Widefield eyepieces WF 10x
(18)

Objectives: A
10x/0.25

A 40x/0.65

FL 50x/1.0 (Oil)

A 100x/1.25 (Oil)

The special
objective FL 50 x/1.0 (Oil) for fluorescence microscopy gives brighter images
and the highest fluorescence intensity.

Filterblock for FITC

Condenser NA 0.9
for brightfield, filterslide.

Can be equipped for
phase contrast and darkfield.

lighting 12 V/50 W or 12
V/100 W. Mechanical stage, quintuple nosepiece.

Widefield eyepieces WF 10x
(18)

Objectives: PL
4x/0.10

PL 10x/0.25

PL 20x/0.40

PL 40x/0.65

PL 100x/1.25 (Oil)

Condenser NA 0.9
for brightfield, filterslide.

Objectives: A 10/0.25

A 20/0.35

A 40/0.65

A 63/0.85

Polarising
equipment with rotatable polariser
and analyzer and compensator for first order red. Centering telescope for conoscopy and condenser NA 1.25.

Binocular laboratory
microscope with variable halogen lighting 6 V/30 W, quadruple reversed
nosepiece, allowing access to the specimen. Widefield
eyepieces WF 10 x (18).

Objectives: A 4x/0.10

A 10x/0.25

A 40x/0.65

A 100x/1.25 (Oil)

Condenser NA 0,9 for brightfield, filterslide.

H 600 WILOCYT

FOR THE
GYNAECOLOGY LABORATORY (FOR

STAINED AND
UNSTAINED CELL PREPARATIONS)

Binocular
laboratory microscope with variable halogen lighting 6 V/30 W, quadruple
reversed nosepiece, allowing easier access to the specimen. Widefield eyepieces

WF 10 x (18).

Objectives: A
10x/0.25

APh 20x/0.35

APh 40x/0.65

A 100x/1.25 (Oil)

Condenser NA 0.9
for brightfield, darkfield
and phase contrast (up to 40 x), filterslide.

H 600 PHACO

As H 600 Wilocyt, but with quintuple nosepiece and five semiplanachromat phase contrast objectives, for unstained
specimen preparations examined under phase contrast.

OBJECTIVES

Ranges of objectives
for transmitted light investigations:

– Achromatic

– Semiplan achromatic

–
Plan-C-achromatic

– Plan
achromatic

– Phase
contrast achromatic and semiplanachromatic

– Infinity
corrected

EYEPIECES

Widefield eyepieces,
standard and for spectacle wearers are available in the following
magnifications: 10 x, 12,5 x. For measuring
eyepieces fitted with a graticule are available.

BINOCULAR
PHOTOTUBES

Binocular phototube
70/30

Light is split with
70% to the phototube and attached device eg video camera,
30 % to the eyepieces enabling simultaneous observation.

Binocular
phototube 100/100

100 % of the
light is directed either to the eyepieces or the phototube. This system is
most useful for low light intensity situations, as all the available light is
directed to the photographic device

Video cameras are
fitted to the phototube by means of a C mount.

DISCUSSION TUBE
DB 2

For the
simultaneous viewing of a specimen by two observers a discussion tube
equipped with a pointer is available.

ACCESSORIES FOR
FLUORESCENCE FILTER BLOCK

Filtersets for
H 600 AFL & H 600 LL AFL

Filterset A
(DAPI) Excitation range 340 - 380 nm,

Dichroic
mirror 400 nm, suppression filter 435 - 485 nm

Filterset B
Excitation range 355 - 425 nm,

Dichroic
mirror 440 nm, suppression filter 455 nm

Filterset C
Excitation range 436/7 nm

Dichroic
mirror 475 nm, suppression filter 495 nm

Filterset D
(FITC) Excitation range 450 - 490 nm,

Dichroic
mirror 510 nm, suppression filter 520 nm

Filterset E
(TRITC) Excitation range 515 - 560 nm,

Dichroic
mirror 580 nm, suppression filter 580 nm

Filterset F
Excitation range 355 - 470 nm,

Dichroic
mirror 510 nm, suppression filter 520 nm

Filterset G
Excitation range 355 - 425 nm,

Dichroic
mirror 400 nm, suppression filter 455 nm

Filterset Texas Red Excitation
range 540 - 580 nm

Dichroic
mirror 595 nm, suppression filter 600 - 660 nm

Filterset Rhodamin (red version) Excitation range 546 nm,

Dichroic
mirror 565 nm, suppression filter 600 - 660 nm

Further filterblocks
upon request

Incorporation of further filtersets
(per filterset)

Filterslider for
fluorescence